Use of this information is subject to copyright laws and may require the permission of the owner of the information, as described in the ECHA Legal Notice.
EC number: 230-603-6
CAS number: 7216-56-0
assigned to the boiling of the substance were observed and measurement
was stopped at 235 °C.
corresponding to these peaks were used to create the following
Initial temperature (°C)
Final Temperature (°C)
Heating rate (°C/min)
Liquid nitrogen cooling
injection rate in the furnace.
taken: 13.03 mg
Melting point is observed at -29.9 °C.
pressure: 984.0 hPa
taken: 10.51 mg
point was observed at -30.4 °C.
study was performed to determine the melting point/range of test item
(E,Z)-2,6-DIMETHYLOCTA-2,4,6-TRIENE. The method used was Differential
Scanning Calorimetry (DSC) and the test was according to Regulation EC
No. 440/2008 Method A.1 and A.2. and OECD Test Guideline 102 , "Melting
this method, two
crucibles were put in the apparatus furnace under inert atmosphere
(helium) at atmospheric pressure. One crucible contained the test item,
the other was empty and considered as the reference. A specific
temperature program was applied. The crucibles were linked to two
thermocouples fixed on the TG-DSC sample carrier. Heat
flow differences between the two crucibles and mass variation were
recorded on a thermogram.
cooling system using nitrogen allowed experiments with low temperature
down to -150°C.
trials were conducted and corresponding thermograms were obtained,
giving us the following results:
melting point values respect validity criteria (less than 0.5°C of
difference up to 326.85 °C).
the melting point of the Indium
respects the checking cards limits (156.5 °C for a reference value of
156.6 °C) so as its enthalpy (28.8
J/g for a reference value of 28.6 J/g).
No melting point of substance (E,Z)-2,6
-DIMETHYLOCTA-2,4,6 -TRIENE was observed from - 20°C. The melting point
of the substance is -30.1 °C.
melting point of substance (E,Z)-DIMETHYLOCTA-2,4,6-TRIENE is - 30.1°C.
Key study conducted according to the OECD
102 Guideline and the EU A.1 method, under GLP without deviation and
with certificate of analysis included. Therefore, a reliability of 1 is
assigned and the result of this key study is proposed as the end point
Information on Registered Substances comes from registration dossiers which have been assigned a registration number. The assignment of a registration number does however not guarantee that the information in the dossier is correct or that the dossier is compliant with Regulation (EC) No 1907/2006 (the REACH Regulation). This information has not been reviewed or verified by the Agency or any other authority. The content is subject to change without prior notice.Reproduction or further distribution of this information may be subject to copyright protection. Use of the information without obtaining the permission from the owner(s) of the respective information might violate the rights of the owner.
Welcome to the ECHA website. This site is not fully supported in Internet Explorer 7 (and earlier versions). Please upgrade your Internet Explorer to a newer version.
Do not show this message again